One unbundle function wired to the Build Waveform function is all that you need. The cluster used for the spectrum output is just an older method that predates the waveform data type and it is what is shown for the Graph help. Although theoretically I can plot it by forming the x-axis myself in excel knowing the initial value and the step size , but because I’m going to collect a great amount of experiments, this is too tedious to do. Is there a way to fix this? As shown in the picture below, the output cluster doesn’t have the corresponding “t0” asked by the Build Waveform function. I’ve tried using an other the control board E Type to filter by text Filter by tag Sort Sort by date created:
|Date Added:||15 December 2010|
|File Size:||63.75 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
How to save plots from Agilent EA spectrum analyzer. Labvlew doesn’t appear that t Hello there, I am using EA spectrum analyzer to do noise figure measurement.
ALCIOM – design house company- specialized in mixed signal electronics
Type to filter by text Filter by tag Sort Sort by date created: Thank you so much for your input! Michael last modified by mriisj Hi, Like anyone knows, there is no schematic available for power supply Was there ever a service manual made for the A?
I was also trying to modify the output from “Trace Read” function in its driver library.
Then I loaded those data and ploted them. Message 10 of One unbundle function wired to the Build Waveform function is all that you need.
HP 8594E battery replacement procedure
I also need the x da As shown in the picture below, the output cluster doesn’t have the corresponding “t0” asked by the Build Waveform function. Get a feed of this content Use this view labvview a tile. There are three different boardsin the poqer supply: Message 5 of Because the “Trace Read” palate returns only 1 array plus the initial point and step size they are both frequencies hp8594ee this casewhat I end up having in the saved spreadsheet was 2 numbers plus 1 array of data the amplitudes.
How to save plots from Agilent EA spectrum analyzer Solved! I opened the measured data sdf-format in VSA and resave it as trace and recording in mat-format.
What is the best way or methods including equipment setup, etc to measure the noise figure at low frequency accurately say from 1KHz to 10MHz? Message 4 of Questions All questions Open questions Answered questions Questions with helpful answers.
I really want to learn how hp859e automatically save data in spreadsheet format out of the plot. I use the analyzer NA.
GPIB Example for LabVIEW – Discussion Forums – National Instruments
When the SA have the correct spectrum plot shown in the “Trace Waveform” window in the front panel, I gp8594e know how to set the program up so that it stores the plot from the window screen into spreadsheet format automatically.
While I was working on it today, I realized a problem. Thanks a lot for your help!